Angle-Resolved Photo-Emission Spectroscopy with a Laser
Laser-based ARPES (Angle-Resolved Photo-Emission Spectroscopy) is a form of ARPES that uses a laser as an excitation source. Laser-based ARPES is operated with a narrow bandwidth UV laser source and offers several advantages over synchrotron radiation: A laser fits on a table (ok… a large table), and experiments can be performed at any time and independently of rare beam times in synchrotron facilities.
More technically speaking, a narrow-band UV laser offers a contantly high photon flux with an extremely high energy resolution of up to 0.1 meV. For comparison, the energy resolution of synchrotron radiation is typically in the range of 5 – 20 meV – and is therefore worse than the resolution of a good energy analyzer with about 1 meV.
The photon energy of the laser source can be varied between 5.77 eV and 6.5 eV. As these are low-energy photons, the Angle-Resolved Photo-Emission Spectroscopy benefits because the bulk sensitivity is significantly enhanced and the momentum resolution increased (compared to synchrotron radiation).
Low-energy photons are sufficient to induce photoemission in many samples. However, it should be noted that due to the lower photon energy compared to synchrotron radiation, Laser-ARPES is particularly suitable for the measurement of electronic states close to the Fermi level.
Laser ARPES/ARUPS with a High-Power UV Source
Tunable deep-UV pulses from 190 … 215 nm / 6.5 … 5.77 eV are available with APE’s ARPES-laser source, constiting of a HarmoniXX FHG in combination with Levante Emerald HP.
The energy-tunable system provides high laser power in the range of several mW. Its brillant and monochromatic laser light is ideally suited for high-resolution ARPES (Angle-resolved Photoemission Spectroscopy) or ARUPS (Angle-resolved Ultraviolet Photoemission Spectroscopy).
- Tunable (Vacuum)-UV from 190 … 215 nm / 6.5 … 5.77 eV
- Extremely narrow spectral bandwidth of < 0.2 meV
- High photon flux of > 10^15 photons/s
- Exceptional beam quality of M2 < 1.2
- Automated wavelength tuning via software
- Ideally suited for table-top laser-based ARPES / ARUPS
Laser ARPES/ARUPS with a Narrow Spectral Bandwidth Down to 0.1 meV
ARPES on Graphene/Ir(111)
The following measurement were performed with APE’s 190 nm / 6.5 VUV system. The ARPES measurement were taken on a SPECS GmbH system with a PHOIBOS 150 hemispherical analyzer and a 2D-CCD detector.
ARPES maps of graphene/Ir(111) sample measured with laser around normal emission geometry showing the Rashbasplit of graphene/Ir(111) surface state. Two different entrance slits of the Electron Analyzer were used, resulting in different energy resolution (marked in the figure).
hv = 6.31 eV
Lens mode: WAM
Nr of energy points: 251 (Sweep mode)
Nr of 0-angle channels: 300
Dependence on Temperature. 3D ARPES data sets for graphene/Ir(111) sample measured with laser (lens mode: WAM). Measurements were performed at T=300 K (RT, see image) and T=120 K (LT, not shown here).
polar β-scan around NE
hv = 6.31 eV
Nr of energy points: 251
Nr of Θ-angle channels: 300
Nr of β-angle steps: 101
Total acquisition time: 50 min
Turning Complexity into a Turn-Key System
This light source enables spectroscopy and pump probe measurements in the deep UV range < 200 nm. The system consists of the following sub-components:
Creating a 190 nm / 6.5 eV Table-top System
The HarmoniXX allows access to the wavelength range of 190 … 215 nm (6.5 … 5.77 eV) by mixing the THG with a fundamental wavelength to create the fourth harmonic. The harmonic pulses are of narrow bandwidth nature, enabling a high resolution across the entire tuning range.
Paladin 532 – 20000 from Coherent Inc. is ideal for pumping the OPO. It is enabling a wide tuning range and high-average powers of the OPO output. Of course, other pump lasers can be incorporated into the setup as well.
|Levante Emerald HP + HarmoniXX FHG (incl. THG, SHG)|
|Wavelength Range||< 192 ... 215 nm | > 6.5 ... 5.77 eV (FHG)|
|< 240 ... 320 nm | > 5.12 ... 3.87 eV (THG)|
|< 360 ... 480 nm | > 3.44 ... 2.58 eV (SHG)|
|< 690 ... 990 nm (Levante Emerald HP OPO Signal)|
|Power Output||> 11 mW at 207 nm | 6 eV (FHG)|
|> 75 mW at 280 nm | 4.4 eV (THG)|
|> 410 mW at 400 nm | 1.55 eV (SHG; THG components removed)|
|> 6000 mW at 760 ... 960 nm (OPO Signal)|
|Pulse Width||approx. 15 ps|
|Spectral Bandwidth FHG||5 pm | 0.16 meV at 196 nm | 6.46 eV|
|Repetition Rate||80 MHz|
|M2||< 1.2 (typical 1.1)|
|Polarization||Linear / Horizontal|
|Power||> 20 W|
|Pulse Width||Approx. 15 ps|
|Manufacturer||Coherent Inc. (Paladin 532 -20000 )|
|Remote Control||Possible via USB / Ethernet TCP/IP / Serial RS232|
Datasheets & Brochures
APE Laser ARPES 190 nm 6.5 eV UV Rev.3.1.1 (english / pdf)
APE Laser ARPES White Paper Rev.4.0.0 (english / pdf)
Email & Phone Contacts
APE has distributors around the world to give you the best support. Choose a country to find your local sales contact: